HANDLING THIS SERVICE MANUAL About This Service Manual n. Scope This Service Manual applies to the Fuji Medical Dry Laser Imager DRYPIX Service Technicians PACS / DR / CR / Digital Forum Drypix Service Manual CARLOS RUEDA Innovacion Tecnologica Hello, I am trying. View and Download FujiFilm DRYPIX service manual online. DRYPIX Printer pdf manual download.
|Genre:||Health and Food|
|Published (Last):||10 December 2013|
|PDF File Size:||11.80 Mb|
|ePub File Size:||1.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
Don’t show me this message again. Remove the filter section cooling fan. MC Remove the screws securing the sub-scanning unit.
service manual – Fujifilm DryPix | MedWrench
When the data transfer is completed, the [File transfer is completed. Enter text from picture: Tray kanual assembly Back side FP7C Page MC 7.
Removing Procedure Insert the shutter into the film tray. Page 46 MC Remove the suction cup arm. Page 3 The screws painted in yellow must be adjusted after the installation of parts.
DryPix Plus (4000) QC for Mammography
DryPix Operation Manual. Page 84 Remove the sub-scanning unit fixing bracket.
However, there mxnual no need to pull out the rail pushed into the equipment. Page 58 Secure the gear while pressing the flange of the idle gear against the gears in the arrow direction shown below. Remove the film positioning assembly lower. EPS Reinstall in the reverse order of removal.
You can add Community Subscriptions in the search bar that says “Subscribe to more communities Page 18 Reinstalling Procedure Reinstall in the reverse order of removal.
My Bench Order History Sign out. MC If the measured values are not the specified values, loosen the screws securing mznual sub-scanning unit. If so, circle the out of control data points, correct the cause of the problem refer sefvice the Performance and Corrective Action section of this document and repeat the test. Page 94 26 x 36 cm size 20 x 25 cm size It accommodates all common film sizes, and has an optional second film drawer and top-mounted msnual film sorter.
Page 52 Push down the plate, and with an approximately 50 mm space between the plate and stopper, push the suction cup against the film. Page 56 MC Remove the screws securing the sub-scanning unit.
The DryPix is a dry laser imager which combines the high quality and performance of Fuji’s centralized imagers, with an extremely compact design and convenient volt power, required for decentralized imaging applications. Insert the sub-scanning unit into the equipment, and temporarily secure with the screws. Filter Section Cooling Fan Remove the cooling fan assembly.
Forums Documents and Manuals. Remove the grip position detection sensor servvice. To print the manual completely, please, download it. MC Remove the tray inside cover.
DryPix Plus () QC for Mammography
Page 24 Reinstalling Procedure Reinstall in the reverse order of removal. Geometry Recording Length and Width 4.
The x-ray detector system is fundamental to the performance of Push down the plate, and with an approximately 50 mm space between the plate and stopper, push the suction cup against the film. Page When reinstalling the adiabatic cover, do not sandwich the thermistor cable.
Page 16 MC Remove the tray manuzl cover. Conveyor Unit Guide Remove the conveyance driving roller lower. Page Press the LED cover of the density measurement light-emitting board against the position shown in the figure below, and secure.